Mechanical Level Transmitter
The buoyancy generated by the float operate the spring. And the gear connected to the spring moves to display the liquid level. In order to export current signal, you need to mount potentiometer RI convertor on the gear.
Part Name and Dimension
① Supply power to the product. (DC 24V).
② Pull down the float to the bottom point and the indicator shows 0% point.
③ If it is out of 0% range, turn the zero volume on the convertor to set as exact “0”.
④ After “zero” adjustment, pull up the float to the top point and the indicator shows 100% point.
⑤ If it is out of 100% range, turn the span volume on the convertor to set as exact “100”.
⑥ Repeat the ①~⑤ step 2-3 times, and install the sensor in the tank.